GEM-based detector Leaflet:
|GEM-based detector Manual:|
|GEM HV divider Description:|
|Perfect for cosmic radiation detection and GEM R&D, the detector is reliable and easy to use.|
TTA Techtra is pleased to offer GEM-based detector kits which include:
- A 100 x 100 mm2 standard detector kit compatible with the CERN kits
- Front-end readout electronics
- Power supplies and gas armature
The detector kit from TTA Techtra allows for easy entry into the GEM research and development.
Detector Gas-box and Hardware
TTA Techtra offers assembled and validated GEM detectors compatible with the CERN-made detector kits and pad planes. The detector contains a pad plane (Figure 1), 3 GEM foils, and a drift electrode. The GEM foils have special snap-on connectors which permit GEM foil removal and installation without soldering (Figure 2). A resistive voltage-divider plugs onto the readout board to polarize the GEM foils and the drift plane (Figure 3).
To operate, the detector needs a 5kV high voltage power supply and a source of working gas, typically a mixture of Argon and CO2. Compatible power supplies add the gas armature is available separately upon request.
|Figure 1. Close-up of the pad-plane with perpendicular XY strips.|
|Figure 2. GEM foils are connected using snap-on connectors for easy foil exchange|
|Figure 3. The L-shaped voltage divider plugs onto the pad plane and polarizes all GEM foils and the drift electrode.|
The L-shaped readout board is pictured in Figure 4. It plugs directly into the Panasonic connectors on the pad plane. It can concurrently read 256 or 512 channels. Each channel is digitized using a switched integrator that guarantees a loss-less collection of the input current (Figure 5). The readout electronics are based on the DDC264 chips. Each channel can be sampled every 160μs with a 20 bit accuracy, which is perfect to read cosmic radiation. Moreover, the readout board has a controlling FPGA device and an Ethernet controller. We provide dedicated software for the PC that controls device’s operation.
|Figure 4. Readout electronics. The board plugs directly onto the pad plane. It connects to the PC via Ethernet|
|Figure 5. Single channel readout architecture|